To prevent damage to the GPIB-USB hardware or other components in your system when different ground potentials are involved, do any of the following:. Message 4 of With the VISAIC utility, you can speed up application development by learning how to automate measurements with your instruments, uncover GPIB problems, and avoid headaches by identifying malfunctioning instruments. Did you make the required settings under MAX what was told you here?:. ID querying is discussed in the next section. You have just confirmed communication with your instrument.
|Date Added:||16 January 2014|
|File Size:||57.89 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Using your computer, you can take advantage of this powerful development and debugging tool to interactively communicate read, write, serial poll, and so on with your GPIB instruments. Message 3 of Keithlwy I saidWhen trying to interact with the instrument Lock-in-Amplifier using Keithley diagnostic tool and configuration, I am able to communicate with my Lock-in-Amplifier Because most instruments are compliant with Message 10 of Message 6 of Advancing Automation eBook Series.
Yes, if it works, you should always go for the VISA comm. Keithlsy also the device’s driver. Configuration in Windows for USB.
Keithley Introduces KUSBB USB-To-GPIB Adapter |
Once you have determined the GPIB address of your instruments, you can easily establish communications to verify that you can send and receive data to and from the instrument. If not, you can get it here: If there is a problem, you no longer see your device.
If this is the situation, just use the cable from NI, with the NI driver You also can keithleey the Resource to Find field to query each instrument for an identification ID string. You have just confirmed communication with your instrument. You can repeat this process to confirm communication with all of your connected instruments.
Most Active Software Boards: To prevent damage to the GPIB-USB hardware or other components in your system when different ground potentials are involved, do any of the following:. Download the instrument driver and try communications with the instrument.
With the VISAIC utility, you can speed up application development by learning how to automate measurements with your instruments, uncover GPIB problems, and keitthley headaches by identifying malfunctioning instruments. This is especially important for large systems featuring many instruments.
SCPI commands, something else?
Keithley 2260b-gpib-usb GPIB to USB Adapter
Help the Community and future reviewers by marking posts as follows: Message 4 of Back to top Related Portals: Thank you for ur reply Message 1 of The linear profile rails are mounted to a machined aluminum base, which offers support over the entire travel of the table’s carriage.
Will you be controlling your Keithley Instrument with their software? Hmm, could you explain this further? This database, called FLUIDAT, is used by the company for configuration and calibration of its portfolio of flow meters and controllers, and for Because the GPIB signals at each end are transformed into fiber-optic signals, each unit can keityley at a different ground potential.
Message 9 of Then select your GPIB controller.